How to Start a Vacuum Test:
Press the Menu button (Top Right) Select and start the Vacuum Test cycle from the “Test cycles” submenu.
Close the door and hit the start button
The Vacuum Test program is run with the sterilization chamber empty. The Vacuum test cycle MUST be run on a cold chamber so it should be your first cycle of the day.
If the test fails, the cycle is automatically aboarted and a alarm message is showen. If this occurs contact ATS immediately.
The duration of the test is about 20 minutes